Water Pump capacity 250LPM ,Head 30Mtr, Pipe line Bore40mm,
my DP Transmitter 5000mmH2O. How can i calculate Orifice plate bore dia.
ORIFICE PLATE CALCULATION
Re: ORIFICE PLATE CALCULATION
Here is the report for orifice calculation for you data:
Orifice plate calculator CALCULATION REPORT
1. diameter (D1): D1 = 40 mm
2. diameter (D2): D2 = 5.55 mm
3. pressure on bigger diameter (p1): p1 = 10328.746 mmWS
4. pressure on smaller diameter (p2): p2 = 52274.996 Pa
5. pressure drop (p1p2): p1p2 = 5000 mmWS
6. kinematic viscosity (ν): ν = 1.006 mm2/s
7. dynamic viscosity (μ): μ = 1.2332191E006 Pas
8. density (ρ): ρ = 1.2258638 kg/m3
9. volumetric flow rate (q): q = 248.452 l/min
10. mass flow rate (w): w = 0.0050761485 kg/s
11. velocity (V1): V1 = 3.295203 m/s
12. velocity (V2): V2 = 171.16548 m/s
13. Reynolds number (ReD): ReD = 131021.99
14. coefficient of discharge (Cd): Cd = 0.6054161
Use calculator at:
http://www.pipeflowcalculations.com/orifice/
and select liquid as flowing fluid
So your answer is 5.5 mm bore will create 5000 mmWS pressure drop with 250 l/m of flowing water.
Orifice plate calculator CALCULATION REPORT
1. diameter (D1): D1 = 40 mm
2. diameter (D2): D2 = 5.55 mm
3. pressure on bigger diameter (p1): p1 = 10328.746 mmWS
4. pressure on smaller diameter (p2): p2 = 52274.996 Pa
5. pressure drop (p1p2): p1p2 = 5000 mmWS
6. kinematic viscosity (ν): ν = 1.006 mm2/s
7. dynamic viscosity (μ): μ = 1.2332191E006 Pas
8. density (ρ): ρ = 1.2258638 kg/m3
9. volumetric flow rate (q): q = 248.452 l/min
10. mass flow rate (w): w = 0.0050761485 kg/s
11. velocity (V1): V1 = 3.295203 m/s
12. velocity (V2): V2 = 171.16548 m/s
13. Reynolds number (ReD): ReD = 131021.99
14. coefficient of discharge (Cd): Cd = 0.6054161
Use calculator at:
http://www.pipeflowcalculations.com/orifice/
and select liquid as flowing fluid
So your answer is 5.5 mm bore will create 5000 mmWS pressure drop with 250 l/m of flowing water.
Pipe flow calculations  since 2000
Re: ORIFICE PLATE CALCULATION
can we use same parameters for calculating the ORIFICE PLATE for gas line also
please suggest
Ajay
please suggest
Ajay
Re: ORIFICE PLATE CALCULATION
Yes, as expansion factor Y is calculated as given in theory page:
http://www.pipeflowcalculations.com/pip ... ifices.htm
http://www.pipeflowcalculations.com/pip ... ifices.htm
Pipe flow calculations  since 2000

 Posts:1
 Joined:Sat Nov 03, 2012 12:56 pm
Re: ORIFICE PLATE CALCULATION
I have pipe dia 4 inch, line pressure 3kg/cm^2 and i have DP transmitter with range of 0 to 10MPa so WHAT WILL BE diA OF ORIFICE PLATE FOR NATURAL GAS FLOW MEASUREMENT
Re: ORIFICE PLATE CALCULATION
What flow rates do you expect to have?
Pipe flow calculations  since 2000

 Posts:1
 Joined:Fri Dec 14, 2018 5:15 pm
Re: ORIFICE PLATE CALCULATION
What will be the orifice diameter for steam line as per following data
Line size 4 Inch
Inlet pressure 410 kg/cm2
Temp 200250°c
Flow rate 510 TPH
Line size 4 Inch
Inlet pressure 410 kg/cm2
Temp 200250°c
Flow rate 510 TPH